Where can I get the WFM601 (not A+ E+ M or i) calibration software?
Click the link below to download the WFM601 calibration software which normally comes on a floppy disk bundled with the service manual for these instruments. Once downloaded, unzip this archive and copy the contents of it to a floppy disk. Then run the CAL.EXE program from that floppy disk on a PC running DOS. From here the software and the WFM601 service manual should walk you through the necessary steps for calibrating your instrument.
NOTE: This software is intended for WFM601's that do not have any letters at the end of their model numbers. The WFM601A, E, M and i models use different calibration software.
NOTE: You will need to run this software on a PC that is running true MS-DOS. Simply running the software in a DOS command windows within Windows will not work because the tool needs access to hardware ports which are protected under Windows operating systems.
FAQ ID 62366すべてのFAQを表示 »